Instructions on posting Experiment Authorization and Experiment Hazard Control Plan forms
MCR Operators are not allowed to post ESAFs during the 2018-01 Run. If you get a call in the off-shift to post an ESAF, please let the user know you are not allowed to post an ESAF and the posting and start of the experiment will have to wait until the on-duty Floor Coordinator arrives at 0800 the following morning.
Instructions on how to reset Trips of the Personnel Safety System
All beamline Personnel Safety Systems have been fully upgraded by the SI Group. Please treat "Trips" as the former "Minor Faults," and reset the trip up to three times per shift with the PSS Minor Fault key (MIF). When there is a "Trip," either the Chain A or Chain B red LED is steady lit on the PSS panel, and when the "Trip" is reset, the red LED will be off. If a trip occurs more than three times per shift please call the on-duty Floor Coordinator. Treat "Faults" at all beamlines as the former "Serious/Major Faults," and call the on-duty Floor Coordinator.